             Retention Time Locking Data Report

  Retention Locked Method: D:\MassHunter\GCMS\1\methods\1ul_Feihn_10_1splitM.M
Retention Locked Cal Date:  
               Instrument:  
                 Operator:  

Method Lock is currently On

Compound: Myristic acid d27 02102020

Retention Time Calibration:

               ml/min      Time     Spec   Deviation
    File        Flow       min.     Xcor    Seconds

  RTLOCK1.D     0.47     17.449     -NA-     24.162
  RTLOCK2.D     0.52     17.234     -NA-     11.250
  RTLOCK3.D     0.58     17.046     -NA-      0.000
  RTLOCK4.D     0.64     16.879     -NA-     -9.996
  RTLOCK5.D     0.70     16.731     -NA-     -18.894

   Maximum Deviation: 24.162 seconds
           RTL Curve: R = 0.111e0 A*A - 4.121e0 A + 3.854e1

   Terms of Curve Fit:
       Constant = 3.854e1
         Linear = -4.121e0
      Quadratic = 0.111e0
    Coefficient = 0.999985 ** Good Fit **

Locked Retention Time information:

    Retention Locked File: D:\MassHunter\GCMS\1\methods\1ul_Feihn_10_1splitM.M\RTLOCK\RTLOCK3.D
                 Acq Date: 07 Feb 2020 4:54 pm
               Instrument: GC MSD
                 Operator: Holden

  Measured Retention Time: 17.046  Flow: 0.581 (ml/min)
    Locked Retention Time: 17.046  Flow: 0.581 (ml/min)
   ** Locked RT and Flow Within Calibrated Limits **

   Lock run spectrum XCor: 1.0000


 Report created: Mon Feb 10 10:46:42 2020
