<MSAcqMethod xmlns:xsi="http://www.w3.org/2001/XMLSchema-instance" xsi:noNamespaceSchemaLocation="qqqacqmethod.xsd">
	<version>2012002</version>
	<msInstrument>QQQ</msInstrument>
	<ionSource>XESI</ionSource>
	<tuneFile>D:\MassHunter\Tune\QQQ\G6470A\atunes.TUNE.XML</tuneFile>
	<stopMode>ByPumpTime</stopMode>
	<stopTime>1</stopTime>
	<collisionGasOn>true</collisionGasOn>
	<APPILampOn>true</APPILampOn>
	<isTimeFilterEnabled>true</isTimeFilterEnabled>
	<timeFilterPeakWidth>0.07</timeFilterPeakWidth>
	<timeSegments>
		<timeSegment><index>1</index>
			<startTime>0</startTime>
			<diverterValveState>ToMS</diverterValveState>
			<sourceParameters>
				<sourceParameter>
					<id>DGasHeater</id>
					<posPolarityValue>200</posPolarityValue>
					<negPolarityValue>200</negPolarityValue>
				</sourceParameter>
				<sourceParameter>
					<id>DGasFlow</id>
					<posPolarityValue>11</posPolarityValue>
					<negPolarityValue>11</negPolarityValue>
				</sourceParameter>
				<sourceParameter>
					<id>NebulizerPressure</id>
					<posPolarityValue>40</posPolarityValue>
					<negPolarityValue>40</negPolarityValue>
				</sourceParameter>
				<sourceParameter>
					<id>SheathGasHeater</id>
					<posPolarityValue>400</posPolarityValue>
					<negPolarityValue>400</negPolarityValue>
				</sourceParameter>
				<sourceParameter>
					<id>SheathGasFlow</id>
					<posPolarityValue>12</posPolarityValue>
					<negPolarityValue>12</negPolarityValue>
				</sourceParameter>
				<sourceParameter>
					<id>VCap</id>
					<posPolarityValue>3000</posPolarityValue>
					<negPolarityValue>3000</negPolarityValue>
				</sourceParameter>
				<sourceParameter>
					<id>VCharging</id>
					<posPolarityValue>500</posPolarityValue>
					<negPolarityValue>0</negPolarityValue>
				</sourceParameter>
			</sourceParameters>
			<isDataSaved>true</isDataSaved>
			<scanSegments>
				<scanSegment><index>1</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>8.2</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>2PG</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>185</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>97</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>150</fragmentor>
							<collisionEnergy>15</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>2PG</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>2</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>8.2</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>3PG</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>185</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>97</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>150</fragmentor>
							<collisionEnergy>15</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>3PG</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>3</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>8.6</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>6PG</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>275</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>97</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>190</fragmentor>
							<collisionEnergy>15</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>6PG</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>4</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>1.5</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>A</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>268.2</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>136.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>100</fragmentor>
							<collisionEnergy>18</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>A</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>5</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>2.1</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>AcAc</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>101.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>57</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>60</fragmentor>
							<collisionEnergy>9</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>AcAc</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>6</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>7.4</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>AcCoA</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>810.2</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>303.3</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>220</fragmentor>
							<collisionEnergy>33</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>AcCoA</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>7</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>7.4</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>AcCoA_neg</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>808.2</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>408</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>260</fragmentor>
							<collisionEnergy>38</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>AcCoA</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>8</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>7.7</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>ADP</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>428.2</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>136</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>128</fragmentor>
							<collisionEnergy>32</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>ADP</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>9</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>6.1</scheduledTime>
					<timeWindow>2</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>aKG</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>145</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>101.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>60</fragmentor>
							<collisionEnergy>5</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>aKG</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>10</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>4.3</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Ala</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>90.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>44.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>35</fragmentor>
							<collisionEnergy>12</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Ala</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>11</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>7</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>AMP</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>348.2</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>136</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>118</fragmentor>
							<collisionEnergy>20</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>AMP</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>12</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>9</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Arg_70</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>175.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>70.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>105</fragmentor>
							<collisionEnergy>29</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Arg</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>13</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>5.2</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Asn</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>133.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>74</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>72</fragmentor>
							<collisionEnergy>16</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Asn</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>14</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>6.6</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Asp</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>132.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>88</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>72</fragmentor>
							<collisionEnergy>12</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Asp</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>15</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>8.1</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>ATP</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>508.2</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>136</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>148</fragmentor>
							<collisionEnergy>36</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>ATP</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>16</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>7.5</scheduledTime>
					<timeWindow>10</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>b-Ala</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>90.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>72.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>80</fragmentor>
							<collisionEnergy>4</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>b-Ala</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>17</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>2.5</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>BHB</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>103</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>59</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>80</fragmentor>
							<collisionEnergy>8</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>BHB</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>18</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>2.1</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>C</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>244.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>112</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>90</fragmentor>
							<collisionEnergy>12</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>C</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>19</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>8.1</scheduledTime>
					<timeWindow>2</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Cit_67</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>191.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>67</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>80</fragmentor>
							<collisionEnergy>26</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Cit/Iso</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>20</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>8.1</scheduledTime>
					<timeWindow>2</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Cit_Iso</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>191.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>111</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>80</fragmentor>
							<collisionEnergy>12</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Cit/Iso</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>21</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>8.1</scheduledTime>
					<timeWindow>2</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Cit_unique</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>191.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>87</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>80</fragmentor>
							<collisionEnergy>17</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Cit/Iso</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>22</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>8.1</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>CMP</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>324.2</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>112</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>97</fragmentor>
							<collisionEnergy>12</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>CMP</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>23</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>8.1</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>CMP</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>324.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>112</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>90</fragmentor>
							<collisionEnergy>16</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>CMP</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>24</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>7.9</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>CMP_neg</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>322.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>79</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>200</fragmentor>
							<collisionEnergy>50</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>CMP</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>25</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>4.4</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Cr</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>132.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>44.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>123</fragmentor>
							<collisionEnergy>24</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Cr</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>26</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>1.7</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Crn</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>114.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>44.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>123</fragmentor>
							<collisionEnergy>20</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Crn</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>27</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>1.2</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>CSA</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>231.3</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>80</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>159</fragmentor>
							<collisionEnergy>36</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>CSA</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>28</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>8.7</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>CTP</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>484.2</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>112</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>118</fragmentor>
							<collisionEnergy>24</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>CTP</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>29</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>4</scheduledTime>
					<timeWindow>3</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Cys</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>122.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>59</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>70</fragmentor>
							<collisionEnergy>28</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Cys</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>30</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>7.7</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>CysCys</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>241.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>74</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>105</fragmentor>
							<collisionEnergy>31</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>CysCys</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>31</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>8</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>dATP</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>492</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>136.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>135</fragmentor>
							<collisionEnergy>20</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>dATP</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>32</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>8.6</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>dCTP</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>468</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>112</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>120</fragmentor>
							<collisionEnergy>14</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>dCTP</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>33</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>8.5</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>dGTP</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>508.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>152.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>135</fragmentor>
							<collisionEnergy>20</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>dGTP</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>34</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>7.7</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>dTTP_pos</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>483</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>81</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>150</fragmentor>
							<collisionEnergy>20</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>dTTP</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>35</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>9.6</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>F16BP</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>339</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>97</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>115</fragmentor>
							<collisionEnergy>21</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>F16BP</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>36</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>7.1</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Fum</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>115</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>71.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>80</fragmentor>
							<collisionEnergy>4</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Fum</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>37</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>2.2</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>G</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>284.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>152.2</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>100</fragmentor>
							<collisionEnergy>12</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>G</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>38</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>7.8</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>G1P_F6P</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>259.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>97</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>200</fragmentor>
							<collisionEnergy>16</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>G1P/F6P</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>39</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>7.8</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>G1P_unique</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>259.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>241</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>200</fragmentor>
							<collisionEnergy>11</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>G1P/F6P</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>40</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>7.5</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>G3P-79</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>171</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>79</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>155</fragmentor>
							<collisionEnergy>17</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>G3P</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>41</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>8.2</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>G6P</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>259.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>97</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>200</fragmentor>
							<collisionEnergy>16</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>G6P</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>42</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>5.1</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Gln</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>147.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>84.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>80</fragmentor>
							<collisionEnergy>18</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Gln</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>43</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>6.6</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Glu_neg</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>146</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>102.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>77</fragmentor>
							<collisionEnergy>12</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Glu</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>44</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>6.6</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Glu_pos</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>148.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>84</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>77</fragmentor>
							<collisionEnergy>16</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Glu</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>45</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>4.9</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Gly</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>76.4</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>30.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>41</fragmentor>
							<collisionEnergy>16</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Gly</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>46</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>7.9</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>GMP</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>364.2</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>152</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>102</fragmentor>
							<collisionEnergy>20</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>GMP</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>47</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>7.9</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>GMP_79</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>362.2</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>79</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>128</fragmentor>
							<collisionEnergy>24</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>GMP</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>48</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>8.7</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>GTP</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>524.2</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>152</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>148</fragmentor>
							<collisionEnergy>24</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>GTP</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>49</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>5.8</scheduledTime>
					<timeWindow>3</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>His</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>156.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>110.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>105</fragmentor>
							<collisionEnergy>14</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>His</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>50</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>1.6</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>HX_neg</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>135.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>92.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>115</fragmentor>
							<collisionEnergy>17</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>HX</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>51</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>1.95</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>I</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>267.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>135.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>145</fragmentor>
							<collisionEnergy>23</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>I</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>52</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>2.4</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Ile</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>132.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>86.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>80</fragmentor>
							<collisionEnergy>9</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Ile</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>53</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>7.6</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>IMP_neg</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>347.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>79</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>140</fragmentor>
							<collisionEnergy>48</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>IMP</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>54</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>7.6</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>IMP_neg135</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>347.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>135</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>140</fragmentor>
							<collisionEnergy>31</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>IMP</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>55</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>7.6</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>IMP_pos</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>349.2</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>137</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>97</fragmentor>
							<collisionEnergy>12</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>IMP</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>56</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>8.1</scheduledTime>
					<timeWindow>2</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Iso_unique</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>191.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>117</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>80</fragmentor>
							<collisionEnergy>14</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Cit/Iso</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>57</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>6.2</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Ita</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>129.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>85.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>71</fragmentor>
							<collisionEnergy>8</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Ita</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>58</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>2.3</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Lac</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>89</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>43.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>41</fragmentor>
							<collisionEnergy>8</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Lac</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>59</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>2.2</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Leu</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>132.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>86.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>80</fragmentor>
							<collisionEnergy>9</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Leu</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>60</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>9.7</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Lys</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>147.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>84.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>80</fragmentor>
							<collisionEnergy>18</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Lys</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>61</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>6.9</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Mal</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>133.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>115</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>80</fragmentor>
							<collisionEnergy>8</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Mal</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>62</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>1.8</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>MES_neg</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>194.2</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>80</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>138</fragmentor>
							<collisionEnergy>32</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>MES</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>63</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>1.8</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>MES_pos</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>196.2</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>100</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>159</fragmentor>
							<collisionEnergy>24</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>MES</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>64</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>2.6</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Met_104</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>150</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>104</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>80</fragmentor>
							<collisionEnergy>8</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Met</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>65</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>2.6</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Met_133</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>150</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>133.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>80</fragmentor>
							<collisionEnergy>7</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Met</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>66</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>5.4</scheduledTime>
					<timeWindow>2</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>MetMal</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>117.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>73.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>60</fragmentor>
							<collisionEnergy>7</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>MetMal</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>67</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>7.1</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>NAD</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>664.4</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>136</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>143</fragmentor>
							<collisionEnergy>56</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>NAD+</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>68</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>6.3</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>NADH</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>666.4</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>649.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>148</fragmentor>
							<collisionEnergy>16</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>NADH</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>69</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>6.3</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>NADH_qual</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>666.4</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>514</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>148</fragmentor>
							<collisionEnergy>24</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>NADH</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>70</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>9.2</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>NADP</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>744.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>136.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>130</fragmentor>
							<collisionEnergy>60</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>NADP</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>71</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>9.2</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>NADP_neg</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>742.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>620</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>130</fragmentor>
							<collisionEnergy>13</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>NADP</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>72</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>8.7</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>NADPH</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>746.2</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>729.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>155</fragmentor>
							<collisionEnergy>17</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>NADPH</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>73</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>8.7</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>NADPH_neg</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>744.2</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>408</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>190</fragmentor>
							<collisionEnergy>37</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>NADPH</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>74</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>8.1</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>PCr</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>212.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>44.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>113</fragmentor>
							<collisionEnergy>36</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>PCr</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>75</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>2.1</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Phe</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>166.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>120.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>90</fragmentor>
							<collisionEnergy>13</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Phe</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>76</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>3.2</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Pro</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>116.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>70</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>72</fragmentor>
							<collisionEnergy>16</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Pro</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>77</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>9.8</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>PRPP_291</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>389</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>291</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>105</fragmentor>
							<collisionEnergy>12</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>PRPP</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>78</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>9.8</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>PRPP_79</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>389</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>79</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>105</fragmentor>
							<collisionEnergy>57</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>PRPP</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>79</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>1.7</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Pyr</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>87</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>43.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>61</fragmentor>
							<collisionEnergy>4</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Pyr</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>80</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>7.6</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>R5P-97</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>229</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>97</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>180</fragmentor>
							<collisionEnergy>12</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>R5P</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>81</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>7.5</scheduledTime>
					<timeWindow>10</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Sarco</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>90.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>44.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>70</fragmentor>
							<collisionEnergy>12</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Sarco</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>82</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>5.2</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Ser</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>106</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>60</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>66</fragmentor>
							<collisionEnergy>12</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Ser</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>83</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>7.1</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Succ</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>117.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>73.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>80</fragmentor>
							<collisionEnergy>10</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Succ</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>84</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>1.3</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>T</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>243.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>126.9</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>80</fragmentor>
							<collisionEnergy>10</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>T</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>85</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>4.4</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Thr</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>120</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>74</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>80</fragmentor>
							<collisionEnergy>9</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Thr</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>86</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>2.3</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Trp_146</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>205.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>146.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>90</fragmentor>
							<collisionEnergy>18</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Trp</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>87</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>2.9</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Tyr_136</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>182</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>136.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>80</fragmentor>
							<collisionEnergy>12</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Tyr</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>88</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>2.9</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Tyr_91</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>182</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>91</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>80</fragmentor>
							<collisionEnergy>32</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Tyr</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>89</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>1.5</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>U</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>243.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>200.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>130</fragmentor>
							<collisionEnergy>8</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>U</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>90</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>6.7</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>UDP-Glc</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>565.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>323.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>165</fragmentor>
							<collisionEnergy>23</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>UDP-Glc</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>91</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>6.7</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>UDP-Glc_79</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>565.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>79</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>165</fragmentor>
							<collisionEnergy>59</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>UDP-Glc</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>92</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>6.7</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>UDP-GlcNAc</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>606.2</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>385</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>160</fragmentor>
							<collisionEnergy>29</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>UDP-GlcNAc</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>93</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>6.7</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>UDP-GlcNAc_79</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>606.2</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>79</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>160</fragmentor>
							<collisionEnergy>58</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>UDP-GlcNAc</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>94</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>7.1</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>UMP</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>325.2</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>97</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>87</fragmentor>
							<collisionEnergy>16</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>UMP</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>95</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>7.1</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>UMP_neg</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>323</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>79</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>210</fragmentor>
							<collisionEnergy>54</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>UMP</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>96</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>7.1</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>UMP_qual</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>325.2</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>41.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>87</fragmentor>
							<collisionEnergy>60</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>UMP</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>97</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Negative</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>8.1</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>UTP</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>483.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>159</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>154</fragmentor>
							<collisionEnergy>36</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>UTP</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
				<scanSegment><index>98</index>
					<ionMode>XESI</ionMode>
					<ionPolarity>Positive</ionPolarity>
					<scanType>DynamicMRM</scanType>
					<dataStorage>PeakDetected</dataStorage>
					<threshold>0</threshold>
					<fragmentorMode>Fixed</fragmentorMode>
					<scheduledTime>3</scheduledTime>
					<timeWindow>1.5</timeWindow>
					<scheduledSetting>MinCycleTime=500</scheduledSetting>
					<isTriggeredMRM>false</isTriggeredMRM>
					<numtMRMRepeats>3</numtMRMRepeats>
					<scanElements>
						<scanElement>
							<index>1</index>
							<compoundName>Val</compoundName>
							<isISTD>false</isISTD>
							<ms1LowMz>118.1</ms1LowMz>
							<ms1Res>Unit</ms1Res>
							<ms2LowMz>55.1</ms2LowMz>
							<ms2Res>Unit</ms2Res>
							<dwell>200</dwell>
							<fragmentor>80</fragmentor>
							<collisionEnergy>10</collisionEnergy>
							<deltaEMV>200</deltaEMV>
							<cellAccVoltage>5</cellAccVoltage>
							<isPrimaryMRM>true</isPrimaryMRM>
							<isTriggerMRM>false</isTriggerMRM>
							<ignoreMRM>false</ignoreMRM>
							<triggerMRMInfo>
								<triggerEntranceDelayTime>0</triggerEntranceDelayTime>
								<triggerDelayTime>0</triggerDelayTime>
								<triggerWindow>0</triggerWindow>
								<isTriggerLogicFlagEnabled>false</isTriggerLogicFlagEnabled>
								<triggerLogicFlag>AND</triggerLogicFlag>
								<triggerRatio>1</triggerRatio>
								<triggerRatioWindow>1</triggerRatioWindow>
								<triggerMRMThreshold>0</triggerMRMThreshold>
							</triggerMRMInfo>
							<funnelRFVoltageHP/>
							<funnelRFVoltageLP/>
							<faimsCV/>
							<faimsDV/>
							<compoundGroup>Val</compoundGroup>
						</scanElement>
					</scanElements>
				</scanSegment>
			</scanSegments>
			<IonFunnel>
			</IonFunnel>
		</timeSegment>
	</timeSegments>
	<instrumentCurves>
		<samplingRate>5</samplingRate>
		<actualID>VCap</actualID>
		<actualID>CapillaryCurrent</actualID>
		<actualID>ChamberCurrent</actualID>
		<actualID>DGasFlow</actualID>
		<actualID>DGasHeater</actualID>
		<actualID>VacuumPressure2</actualID>
		<actualID>MS1Heater</actualID>
		<actualID>MS2Heater</actualID>
		<actualID>NebulizerPressure</actualID>
		<actualID>Pump1Current</actualID>
		<actualID>Pump2Current</actualID>
		<actualID>VacuumPressure1</actualID>
		<actualID>SheathGasFlow</actualID>
		<actualID>SheathGasHeater</actualID>
		<actualID>TurboSpeed1</actualID>
		<actualID>TurboSpeed2</actualID>
	</instrumentCurves>
	<chromatograms>
		<chromatogram>
			<index>1</index>
			<chromType>TIC</chromType>
			<tic/>
			<label>TIC</label>
			<offset>0</offset>
			<yRange>1E+07</yRange>
		</chromatogram>
	</chromatograms>
	<beginDMRMWasteDivValveStartTime>-1</beginDMRMWasteDivValveStartTime>
	<endDMRMWasteDivValveStartTime>-1</endDMRMWasteDivValveStartTime>
</MSAcqMethod>
